digplanet beta 1: Athena
Share digplanet:

Agriculture

Applied sciences

Arts

Belief

Chronology

Culture

Education

Environment

Geography

Health

History

Humanities

Language

Law

Life

Mathematics

Nature

People

Politics

Science

Society

Technology

A simple structure of a dilatometer for the measurement of the thermal expansion of liquids and solids

A dilatometer is a scientific instrument that measures volume changes caused by a physical or chemical process. A familiar application of a dilatometer is the mercury-in-glass thermometer, in which the change in volume of the liquid column is read from a graduated scale. Because mercury has a fairly constant rate of expansion over normal temperature ranges, the volume changes are directly related to temperature.

## Applications

Dilatometers have been used in the fabrication of metallic alloys, compressed and sintered refractory compounds, glasses, ceramic products, composite materials, and plastics.[1]

Dilatometry is also used to monitor the progress of chemical reactions, particularly those displaying a substantial molar volume change (e.g., polymerisation). A specific example is the rate of phase changes.[2]

Another common application of a dilatometer is the measurement of thermal expansion. The thermal expansivity is defined as:

${\displaystyle \alpha ={\frac {1}{V}}{\biggl (}{\frac {\partial V}{\partial T}}{\biggr )}_{p}\ }$

and is an important engineering parameter.

## Types

There are a number of dilatometer types:

• Capacitance dilatometers possess a parallel plate capacitor with a one stationary plate, and one moveable plate. When the sample length changes, it moves the moveable plate, which changes the gap between the plates. The capacitance is inversely proportional to the gap. Changes in length of 10 picometres can be detected.[3]
• Connecting rod (push rod) dilatometer, the sample which can be examined is in the furnace. A connecting rod transfers the thermal expansion to a strain gauge, which measures the shift. Since the measuring system (connecting rod) is exposed to the same temperature as the sample and thereby likewise expands, one obtains a relative value, which must be converted afterwards. Matched low-expansion materials and differential constructions can be used to minimize the influence of connecting rod expansion [4] [5]
• High Resolution - Laser Dilatometer Highest resolution and absolute accuracy is possible with a Michelson Interferometer type Laser Dilatometer. Resolution goes up to picometres. On top the principle of interference measurement give the possibility for much higher accuracy's and it is an absolute measurement technique with no need of calibration.[6][clarification needed]
• Optical dilatometer is an instrument that measures dimension variations of a specimen heated at temperatures that generally range from 25 to 1400 °C. The optical dilatometer allows the monitoring of materials’ expansions and contractions by using a non-contact method: optical group connected to a digital camera captures the images of the expanding/contracting specimen as function of the temperature with a resolution of about ±70 micrometre per pixel.[7] As the system allows to heat up the material and measures its longitudinal/vertical movements without any contact between instrument and specimen, it is possible to analyse the most ductile materials, such as the polymers, as well as the most fragile, such as the incoherent ceramic powders for sintering process.

For simpler measurements in a temperature range from 0 to 100 °C, where water is heated up and flow or over the sample. If linear coefficients of expansion of a metal is to be measured, hot water will running through a pipe made from the metal. The pipe warms up to the temperature of the water and the relative expansion can be determined as a function of the water temperature.

For the measurement of the volumetric expansion of liquids one takes a large glass container filled with water. In an expansion tank (glass container with an accurate volume scale) with the sample liquid. If one heats the water up, the sample liquid expands and the volume changes is read. However the expansion of the sample container must also be taken into consideration.

The expansion and retraction coefficient of gases cannot be measured using dilatometer, since the pressure plays a role here. For such measurements a gas thermometer is more suitable.

Dilatometers often include a mechanism for controlling temperature. This may be a furnace for measurements at elevated temperatures (temperatures to 2000 °C), or a cryostat for measurements at temperatures below room temperature. Metallurgical applications often involve sophisticated temperature controls capable of applying precise temperature-time profiles for heating and quenching the sample.[8]

## References

1. ^ Hans Lehmann, refuge Gatzke Dilatometrie and differential thermal analysis for the evaluation of processes ? ? , 1956.
2. ^ Kastle, J. H.; Kelley, W. P. (July 1904). "On the Rate of Crystallization of Plastic Sulphur". American Chemical Journal 32: 483–503.
3. ^ J. J. Neumeier, R. K. Bollinger, G. E. Timmins, C. R. Lane, R. D. Krogstad, and J. Macaluso, "Capacitive-based dilatometer cell constructed of fused quartz for measuring the thermal expansion of solids", Review of Scientific Instruments 79, 033903 (2008).
4. ^ Theta Industries http://www.theta-us.com/dil/dil1.html
5. ^ http://www.usinenouvelle.com/industry/netzsch-geratebau-gmbh-8585/horizontal-pushrod-dilatometer-dil-pc-p72147.html
6. ^ C.Linseis The next step in dilatometry, invention and use of the Linseis Laser Dilatometer , Linseis Messgeraete GmbH, Selb (Germany)
7. ^ M.Paganelli The Non-contact Optical dilatometer designed for the behaviour of Ceramic Raw Materials , Expert System Solutions S.r.l., Modena (Italy).
8. ^ http://www.theta-us.com/quench/QuenchInst.html

Original courtesy of Wikipedia: http://en.wikipedia.org/wiki/Dilatometer — Please support Wikipedia.
 968 videos foundNext >
 Dilatometer for Coefficient of Thermal Expansion AnalysisC-Therm's New Dilatometer for Thermal Expansion Analysis. Marchetti flat dilatometer DMT - SDMT testMarchetti - Maquesonda workshop lisboa 2013 http://www.marchetti-dmt.it/ DilatometrySee links below... TA Instruments provides the widest range of high precision dilatometers for the measurement of thermal expansion, sintering, phase ... High Precision Dilatometer from C-Therm TechnologiesC-Therm dilatometers offer high resolution and stability across a broad measurement range in the measurement of thermal expansion. With unparalleled ... Dilatometer VideoIntroduction video on the use of the Marchetti Dilatometer. Horizontal Dilatometer Experiment PreparationSubscribe to our channel for helpful tech tips on Thursdays! In this TA TechTip, we discuss how to properly prepare an experiment for the Horizontal Dilatometer ... NETZSCH DIL 402 C DilatometerThe highly vacuum-tight pushrod dilatometer DIL 402 C will meet all of your measurement needs for determining thermal length changes for solids, melts, ... The Flat Dilatometer Test (DMT) in Soil InvestigationsThe DMT is a simple and efficient test for estimating a number of soil engineering properties, and may also be used as an extremely useful logging device since ... NETZSCH DIL402 Expedis DilatometerDIL 402 Expedis Select/Supreme – Horizontal pushrod dilatometer Dilatometry Redefined NanoEye – Largest Measuring Range, Highest Resolution The new, ... Introducing the Horizontal DilatometerThis TA Tech Tip introduces TA Family of Horizontal Dilatometers.
 968 videos foundNext >

We're sorry, but there's no news about "Dilatometer" right now.

 Limit to books that you can completely read online Include partial books (book previews) .gsc-branding { display:block; }

Oops, we seem to be having trouble contacting Twitter